2.laser& 6polar RF
Basal Operation Basal Operation
Laser Function Laser Function
(1)Insert 40K cavitation explode work head to ;
(2)Insert Big or small fat remove work head to ;
(3)Insert Laser X2-1 work head to ;
(4)Insert Laser X2-2 work head to ;
(5)Insert Laser X4-1 work head to ;
(6)Insert Laser X4-2 work head to ;
(7)Insert Laser X4-3 work head to ;
(8)Insert Laser X4-4 work head to ;
(9)Insert Service brake to ;
(10)Insert hex polar RF laser work head to [
2.Insert power supply line to , turn on the
3. As (Fig 1) shows, input default Password “000000” to
enter interface as
4.Click “ENTER...” to enter function interface as
Put the accessories well and connect them to the host.
1.Operation step of Laser X4
(1)Clean the skin, and according the treatment part to
choose the suit band ties and tied them up.
(2)Put the Laser X4 work heads into the band ties, the side
with Led contact to skin directly.
(3)Set intensity of every Laser X4 in interface intensity of laser ......
(4)Setting work time
30 minutes of every treatment part, and click “Time”
1.Applied enough gel on skin need treated.
2.Do not stay unmoved on any part of your body and do not stay on bones.
3.The dai ly treatment should be within 30 minutes .
4.There wi l l be ultrasonic only when you step on the service brake .
6.Do not used in the human body heart, throat and eyel ids. If the skin is sensitive or becomes unhealthy,
the strength of the power should be adjusted down.
Cavitation explode Function Cavitation explode Function
Fat Remove Operation Fat Remove Operation
Mode P1 P2 P3 P4 PE P6
Inspiration 0.3S 0.5S 0.8S -.-
Deflation 0.3S 0.5S 0.4S -.-
1.Click to “Fat remove” in interface (Fig 3) shows, and enter
interface (Fig 7) .
2.Click” ”and ” ”to adjust vacuum intensity,
3.Click “Mode” to choose vacuum mode as follows shows:
t he setting
P1~P3 mode are default mode. If not fit for you,
you can click to a just the setting
/ time is 0.1S~3.0S.
P4~P6 mode can’t adjust / time.
4.Click “Time” set treatment time (about 20 minutes).
5.Appled aroma oil and massage oil on treatment part.
6.Click “ ”to start the function. Click “ ” to stop.
7.The machine will stop automatically if the setting
Mul tipolar RF Operat ion Mul tipolar RF Operat ion
Case of eyes hurt.
1.Click “ Multipolar RF ” as (Fig 3) shows, enter interface as (Fig 8).
2.Set work time, 20 minutes for each part.
3.Click” ”and ” ”to adjust output intensity: 1(weak)~10(strength).
4.Click” ”, the machine is in standby state. Beautician put the service brake under your feet, step on to
start working, move away for pause. Press” ” to stop working.
5.The system default laser is start state, click , the laser function is closed.
7.The machine will pause automatically if the setting time is over.
8.Clean the leftovers with hot towel, use warm towel to Clean the work head.
6.Do not let the laser light straightly shine upon the eyes in case of eyes hurt.
Applied gel on skin needed nursing, the electrode probe move slowly on skin, If there is too much fat, you
can increase the output intensity.
1.Beautician should wear insulated gloves during operation.
2.Do not wear any aglet, both for the beautician and the users.
3.The electrode probe must be fully in contact with skin.Do not contact skin with unipolar when using of
hex polar RF work head otherwise it will produce pulse discharge, gel paste etc. Products like gel maybe
evaporated into a mist.
4.Do not make the RF electrode stay on any part without moving. Keep it moving on skin slowly.
5.hex polar RF work head do not use on eyes and facial.